vFC™ VESICLE FLOW CYTOMETRY EV ANALYSIS ASSAY
FOR COUNTING AND SIZING VESICLES IN BIOFLUIDS, MEDIA, OR BUFFER
$450.00 – $2,100.00
88 in stock
Sample Format: Vesicles from cell culture supernatants or biofluids. Please refer to insructions if analyzing complex samples such as plasma..
Vesicle Size: >70nm (Instrument Dependent)
Certificate of Analysis
Sample Type: Suitable for characterization of extracellular vesicles directly from cell culture supernatant or biofluids such as plasma, urine, or CSF. Can also be used to analyze EVs separated from complex samples using various techniques. See kit manual for sample type specific assay considerations.
Species: Human. Although most reagents are not species specific, certain controls restrict the use of this kit to human samples.
Limits of Detection: vFC™ is suitable for the enumeration of vesicles down to 70nm from background. By following our staining protocols and using vTag™ monoclonal antibodies, surface antigen detection down to 25 molecules per vesicle is possible. LOD is instrument specific. See product manual for additional information.
Sample Storage: Care should be taken to keep assay and storage conditions consistent and report all procedures. If vesicles cannot be analyzed fresh, we recommend storing at -80°C and aliquoting to avoid unnecessary freeze/thaw cycles.
Kit Storage: CAUTION! Multiple storage temperatures required. See product manual for instructions.
What are the limitations of the assay?
vFC™ assay (#CBS4) is intended for the analysis of vesicles from cell culture in supernatant or buffer. It is also useful for analyzing vesicles separated from biofluids. It can detect vesicles down to 70nm on currently available instrumentation. Limit of detection of surface markers per vesicle is approximately 25-30 molecules (depending on the vTag™ antibody used). We have characterized limits for several instruments. Please contact us for instrument specific sensitivity assessments.
Why am I not seeing events from the Lipo100™ control sample?
There are several possible reasons for this issue. First, ensure that the instrument you’re using is suitable for vFC™. Next, make sure that the instrument has been properly set up and qualified. For instrument setup procedures see the instrument setup and configuration document under the datasheets and documentation tab. Finally, double check that samples are being properly diluted. Pay careful attention to details such as the plate being used for dilutions as certain plastics may bind vesicle and significantly impact EV counts. For additional support, chat with us by clicking on the icon at the bottom right of the window.
Why am I seeing high background events?
High number of background events can arise from multiple sources. Ensure proper dilution of samples and only run vesicles that are appropriate for this kit. Also, rarely carryover from samples may lead to background which is especially noticeable in blank samples. This is most commonly observed after running small beads. Run a daily clean cycle prior to continuing analysis. If issues continue, run beads diluted 1:10 in cleaning solution in the future.
Vesicle Flow Cytometry (vFC™) is a standardized assay for reproducible, sensitive characterization of EVs by size, concentration, and cargo
Requirements for Running vFC™
A Suitable Flow Cytometer
The Amnis® CellStream™ is a new cytometer with sensitive optics capable of detecting fluorescence from individual vesicles stained using the vFC™ assay kit. We are currently evaluating the instrument, which appears to have excellent sensitivity for immunofluorescence measurement of EV surface cargo.
The Beckman Coulter Cytoflex™ was the first of the new generation of flow cytometers with suitable sensitivity to detect vesicles by vFC™. The instrument offers more sensitive detection of surface cargo if the yellow laser is activated (which more efficiently excited PE).
The Amnis® ImageStream™ is an imaging cytometer with sensitive optics capable of detecting fluorescence from individual vesicles stained using the vFC™ assay kit. The ImageStream™ has a few differences from conventional flow cytometers that require modifications to the vFC™ protocol. Please refer to the kit instructions for details.
The Novocyte Quanteon™ is a newer instrument that uses the most sensitive silicon photomultiplier (SiPM) detector technology on the market to improve fluorescence sensitivity. Our characterization of this instrument is ongoing.
Suitable Pipette Tips and 96 Well Plates
Appropriate plasticware is required for vFC™. The kit comes with 96 well plates that are compatible with the assay. Most standard plastic is suitable, but all pipette tips, tubes, and plates that are not part of the kit should be tested to ensure they do not interfere with the assay. High binding plastic should be avoided as it will bind vesicles and reduce counts and sensitivity of the assay. Likewise, low retention plastics should also be avoided as many shed particles that result in elevated background.
Principles of vFC™
Characterizing the heterogeneity of vesicles is critical to understanding their biological role. Flow cytometry is the gold standard technology for understanding the heterogeneity of cells and could be incredibly useful for characterizing diverse samples of vesicles except, cells are not vesicles.
Vesicle Flow Cytometry (vFC™) is an assay kit composed of a lipid specific fluorescent probe, synthetic and natural calibrators and controls, and optimized protocols and analysis templates. It is designed to provide a standardized, robust method for EV characterization. vFC™ solves the issues associated with characterizing small lipid particles.
Standardized analysis of data collected using the kit protocol requires use of our data analysis template. This template requires you to download and use the free version 7 of FCS Express Reader. For more information, see our kit handbook.
96 Wells, 5 x 96 Wells
Human, Mouse, Generic
|EV Marker (Antibody) ⓘ||
Tetraspanin (CD9, CD63, CD81) Cocktail [PE], Tetraspanin (CD9, CD63, CD81) Cocktail [PE-Cy7], None